Akvadistillyator PHS Aqua (JSC Tyumensky medical equipment factory and tools) Akvadistillyatora PHS Aqua are intended for production of the distilled water. Akvadistillyatora PHS Aqua are used in medical institutions, drugstores, laboratories; in any production where use of water of the specified extent of cleaning is necessary.
Akvadistillyatora PHS Aqua are intended for production of the distilled water.
The line of akvadistillyator PHS Aqua is presented by the PHS Aqua 4, PHS Aqua 10, PHS Aqua 25 models with a productivity of 4, 10 and 25 liters an hour.
Akvadistillyatora PHS Aqua are used in medical institutions, drugstores, laboratories; in any production where use of water of the specified extent of cleaning is necessary.
Simplicity of operation
Full compliance to all standard requirements
Service support in all regions
Small overall dimensions
Conductivity of the made water of-5x10-4 Cm/m
Protective decorative panels from polished stainless steel
There are no hot surfaces, available to a contact
Unique "fast" electronic sensor of level
Automatic depletion of a chamber of evaporation at termination of work
Easily unite with collections for storage of water in uniform system
Quick-detachable design of a chamber of condensation over an evaporation chamber for more free access and convenience of service and repair
Devices for clarification of steam from drops of water not last distillation
System of gas separation at purification of steam
One point of connection to the drainage line
Time of continuous work – no more than 8 hours
Operation term – not less than 5 years
Automatic shutdown TENOV at the termination of the centralized water supply
Automatic shutdown TENOV at depreciation of water in a chamber of evaporation is lower than admissible
Automatic maintenance of an amount of water, going for evaporation
PARAMETERS: PHS AQUA 4 PHS AQUA 10 PHS AQUA 25
Productivity, l/h 4 10 25
The admission on productivity of-10% of-10% of-10%
Tension, B 220 380 380
Power, kW 3 7,2 16,2
Mass, no more, than a kg 5,5 10,1 18
Time of establishment of an operating mode, no more, min. 8 8 10
Amount of the consumed water, no more, than a l/h 50 120 250