- Country of manufactureRussia
It is intended for the placement of women on it at gynecologic inspections, carrying out different procedures, and also at small surgical interventions.
It is applied in the field of gynecology of medical institutions.
The framework of a chair is made of a steel profile with a polymeric and powder covering, levels of fastening and accessories steel about nickel - a chrome covering, the panel of the basis is made of high - strength plastic, mobile parts of the mechanism of adjustment of height are closed by a removable plastic casing.
The chair is supplied with three electric drives providing adjustment of provisions of back and pelvic sections relatively each other, and if necessary and their joint inclination.
KG - 6 is established in the situation "Trendelenburg".
The pallet (basin) located under a seat plate is made of stainless steel and at change of a tilt angle of section always remains in horizontal position.
At the attached foot panel (an additional complete set) to a seat and completely cast away back the chair forms the uniform plane (table).
On lateral panels of back section and seat levels for fastening of removable parts are placed: an infusional rack, a little table for a hand, an emphasis for hands, kneeboards.
In a casing of back section the holder for one - time hygienic towels is located.
The soft polyurethane foam bed of a chair fitted by a high - quality imitation leather seamless, maintains repeated processing by any solutions applied in medicine, without damage of quality of a product.
The basis of a chair gynecologic is equipped with adjustable support that guarantees a steady arrangement on an uneven surface.
- length - 1380 mm
- width - 580 mm
- seat height - 680 - 980 mm
- section of the directing levels - 27х7 mm
- the allowed loading - not less than 170 kg
- weight - no more than 120 kg
- power supply voltage - 220 V, 50 Hz
- power consumption - no more than 300 VA
- a seat tilt angle - from 0º to + 15º
- a back tilt angle - from - 15º to + 70º.