- Country of manufactureRussia
It is intended for the placement of women on it at gynecologic inspections, carrying out different procedures, and also at small surgical interventions.
It is applied in the field of gynecology of medical institutions.
The framework of a chair gynecologic with the manual KG - 1 drive is executed from a steel profile with a polymeric and powder covering. The basis of a chair is established on a support and two rollers for convenience of movement, supplied with the hinged leaning - back footboard. Elements of fastening of accessories are made of steel with a nikelkhromovy covering. Pillows of a seat and a back of a chair are fitted by an imitation leather.
Dimensions: length (chair/table) - 1122 mm / 1605мм;
length of a seat and back - 1275 mm;
width (with lodgments) - 925 mm;
seat width - 500 mm;
seat height from floor level - 900 mm;
chair height - 1640 mm.
The mass of a chair complete with devices, no more - 75 kg;
rated load on a chair - 120 kg. ;
seat tilt angle from a horizontal from 0 degrees to 20 degrees With, with maximum deviation of ±5 degrees of a back from a vertical - from 20 degrees to 100 degrees With, with maximum deviation of ±5 degrees;
the effort necessary for movement of the pallet (basin or cup), no more - 50(5) N (kgf), the effort necessary for a back rasfiksation at transfer from one situation in another, no more - 120 (12) N (kgf).