Complex X - ray diagnostic medical KMP SIMURGH G4 with accessories.
Configuration of system:
- X - ray diagnostic table
- Digital device
- High - voltage generator
- X - ray tube
Carefully thought over design of a multifunction rotary table of a support allows to consider all variety of diagnostic situations and needs of a wide range of users, doing it ideal for use in such areas as orthopedics, the general X - ray analysis, research gastrointestinal tract with a baric suspension, endoscopy, urology and even an angiography. Big range of the longitudinal movement of a column of a tube the size of the detector of 17х17 inches is provided more, than two - meter coverage without movement of the patient, giving the chance to carry out different procedures in safe conditions.
The best in a class is a detector with a midget size of 139 pixel of micron and the most modern SUREengineAdvance technology allowing to receive unprecedentedly quality images.
The dose decline is reached by system in a complex, thanks to various konstrwuktivnym to features of a long - distance table of a support and functions. It, for example, as a removable lattice, ideal net pulse roentgenoscopy, the new collimator with the multipetal filter of soft beams and virtual collimation. The long - distance rotary table support increases productivity and safety of X - ray diagnostic offices, and the latest appendix accompanying this system will create additional clinical benefits in the competition to other medical institutions.
|X - ray diagnostic table:|
|Table - top||76,5х235 cm|
|The allowed loading||76,5х235 cm|
|Longitudinal range||25 cm|
|Vertikalnyydiapazon||from 47 to 110 cm|
|Longitudinal speed||5 cm/sec.|
|Tilt angle||provision of Trendelenburga 90 °, vertical 90 °.|
|Inclination speed||6 ° / sec.|
|Distance between x - ray focus and a surface of FPD (SID)||from 110 to 150 cm|
|Radiation weed distance between a floor and the center||Vertically - 59,5 to 220 cm
Trendelenburga 46,5 to 207 cm
|High - voltage generator:|
|Two X - ray tubes are connected|
|At a X - ray analysis: tube tension||from 40 to 150 kV; Time – from 1 to 10 sec.|
|At roentgenoscopy: tube tension||from 50 to 125 kV; Tube Current – 0,3 - 20 mavremya roentgenoscopies – 10 min.|
|Three - phase system||with a capacity of 75 kVA and about the power of 80 kV|
|Dimensions of control of an office||700х1900х500 mm|
|Gabaritypaneli of management||308х345х82 mm|
|Lump||no more than 302,5 kg|
|Tension and current of a tube for the maximum electric exit||125 kV and 7,6 ma|
|The maximum tension applicable on a X - ray tube||not less than 150 kV|
|Weeding duration of illumination lighting||continuously 30 sec.|
|Distance between a fitting and focus||not less than 60 mm|
|External dimensions of the collimator||231х259х317 mm|
|Approximate weight||10 kg|
|There are 4 BH filters. The motor drive of a method of the drive is installed|
|Software||MicrosoftWindows 7 OS with the hard drive with a capacity of 500 GB|
|Monitor||17" the inch monitor, with a display 1024х1024 matrix|
|Pulse roentgenoscopy||30/15/7,5/3,25 personnel|
|DSA X - ray analysis||to 7,5 personnel / sec.
|Rated voltage of the power supply||single - phase 200/220/230/240 B, with a frequency – 50/60 Hz.|
|RSM – DSA of a X - ray analysis carry out processing of proofreading in the private domain of the image. SUREengineAdvance allows to receive the quality image. The kept images and data of the patient it is processed in the DICOM format. In the device it is stored at most 80 000 image|
|High - frequency generator|
|Frequency||not less - 50 kHz|
|The complex is equipped with the x - ray generator of different power and specified bukvenno in number of model:
1) P – 20 kW - single - phase system of power supply, with a capacity of 30 kVA;
2) L – 50 kW – one - and three - phase system of power supply, with a capacity from 30 to 50 kVA.
3) V – 65 kW – three - phase system of power supply, with a capacity of 75 kVA;
4) B – 80 kW – three - phase system of power supply, with a capacity of 75 kVA;
When using protective shutdown UZO of invertor type is applied to prevention failures in high - frequency chains. Time of carrying out roentgenoscopy makes 10 min. , and a X - ray analysis fluctuates from 1 ms to 10 pages. At a X - ray analysis tension of a tube makes 40–125 kV, and at roentgenoscopy – 50–125 kV