- BrandАО "Актюбрентген"
- Country of manufactureKazakhstan
The device is intended for medical diagnostic testings by a X - ray analysis method in hospital chambers, accident wards, emergency stations, and also for installation in mobile diagnostic complexes. Provides a possibility of receiving x - ray films of human organs in a standing position, "sitting" and "lying". Positioning is made by movement of a support with a monoblock concerning the patient.
- Microprocessor control the feeding device;
- Mobility of movement in close ward conditions;
- Touch control panel;
- Modern design of the device;
- High quality of receiving pictures;
- The high - frequency feeding device;
- Comfort of procedures of inspection for the patient.
|Parameters of the feeding device:|
|Power of the feeding device, kW||32|
|frequency of forming of a high voltage, kHz||100|
|range of amount of electricity, Mas||0.2 - 200|
|range of change of tension, kV.||40 - 125|
|Maximum current of a tube, ma||425|
|Picture time, sec.||0.3 - 1.5|
|Number of programs of an organavtomatika||240|
|The maximum consumed current, And||16|
|Way of obtaining image||Film|
|Service conditions||Single - phase network|
|network tension, In||220 (±10% )|
|frequency, Hz||50 (±1) Hz|
|resistance of a network, Ohm, no more.||0.5|
|Weight of the device, kg, no more||180|
Completeness: the basis feeding the device, the touch panel
managements, monoblock with a diaphragm and a X - ray tube, a support, the button of pictures, a set of cables.
The software allows to conduct researches both in two - point, and in three - point the modes:
- a method on three points with the choice of kV/mA/mS values;
- a method on two points - according to kV/mAs.
The built - in microprocessor provides automatic calculation of size and duration of exposure, and also for convenience of the user displays these data of the touch display before each picture.
Unlike many other analogs, the device is equipped with system of automatic shutdown after 30 minutes of inaction that allows to keep a resource of a X - ray tube significantly.